To save battery power, if no scanning is detected for 30 seconds, the live imaging screen automatically freezes and the Clarius scanner changes from scanning mode to standby mode. This can be enabled and disabled as necessary.
- Press and hold the freeze icon . The icon will turn green.
- The auto-freeze feature is now disabled. The imaging screen will continue running until you manually freeze it.
- Press and hold until it turns blue.
- Start a new exam.
Note: If you use the scanner continuously for 15 to 20 minutes, you will notice the scanner case heats up. This is normal. Please see article on Managing Heat.