The following table lists the cleaners and disinfectants compatible with your Clarius Ultrasound Scanner and accessories. The products listed in the following table are chemically compatible and have been tested for efficacy.
You may also use products not specifically listed in the compatibility table but with similar active ingredients, as indicated in this list, and marketed for medical use. Because of the large number of available cleaners and disinfectants, it is impossible to have an all-inclusive list. If you are unsure of the suitability of a particular product, go to www.clarius.com/contact and contact Clarius for more information.
|1st Gen. Clarius Scanner||Clarius HD||HD Fan||HD Battery||HD Battery
|Accel® PREVention™ Wipes||LLD, ILD||✔||✔||✔||✔||✔|
|McKesson OPA/28 High-Level Disinfectant Solution||HLD||✔||✔||✔|
|Sani-Cloth® Plus Germicidal Disposable Cloth||LLD, ILD||✔||✔||✔||✔||✔|
|Tristel Trio Wipes System||HLD (a)||✔||✔||✔||✔|
Cleaner & Disinfectant Details
|Solution||Origin (b)||Usage||Active Ingredients|
|Accel® PREVention™ Wipes||CA||Wipe||Hydrogen Peroxide|
|CaviWipes||US||Wipe||Alcohol, Quaternary Ammonia|
|McKesson OPA/28 High-Level Disinfectant Solution||US||Soak||Ortho-phthalaldehyde|
|Sani-Cloth® Plus Germicidal Disposable Cloth||US||Wipe||Alcohol, Quaternary Ammonia|
|Tristel Trio Wipes System||UK||Pre-clean wipe,
|Enzymes, Chlorine Dioxide|
* CL = Cleaner, HLD = High-level disinfectant, ILD = Intermediate-level disinfectant, LLD = Low-level disinfectant, S = Sterilant
a. EU only.
b. AU = Australia, CA = Canada, US = United States, UK = United Kingdom